Datascope Passport XG Multi-Parameter Monitor - Preowned in Very Good Condition
The Datascope Passport XG is a standard in portable and bedside vital signs monitoring. the Passport XG's incredible 52 square-inch display is available in your choice of an Active Matrix Color TFT screen or a bright, high-contrast monochrome electroluminescent screen. Larger NIBP numbers of Automatic Waveform Display make reading your patient's data easier.
Non-invasive blood pressure
Battery back up for transport
21.1 cm x 15.8 cm
Leads I, II, III, aVR, aVL, aVF and V.
Defibrillator overload protection: Withstand Lown and trapezoidal or Edmark waveform up to 400 joules applied to 100 ohms load and electrode at 20 second intervals.
Recovery time: time for trace to return to screen from defibrillator moo pulse is 3 seconds maximum.
Range to digital display; systolic/diastolic mean -20 to +300 mmHg.
Accuracy: 2 mmHg or 2%, whichever is greater to numerical display position.
Zero range: ±120 mmHg minimum at nominal gain.
Range - 30-300 bpm for adult/child, 30-350 bpm for neonatal.
Accuracy - 30-250 ±3 bpm or ±3%, whichever is greater.
Systolic range: 50 to 250 mmHg for adult/ped mode, 15 to 150 mmHg for neonatal mode.
Diastolic range: 30 to 200 mmHg for adult/ped mode, 10 to 150 mmHg for neonatal mode.
Systolic accurancy: mean error less than 7 mmHg, standard deviation less than 8 mmHg (adult/ped), mean error less than 4 mmHg, standard deviation less than 13 mmHg (neonatal)).
Diastolic accuracy: mean error less than 7 mmHg, standard deviation less than 8 mmHg (adult/ped), mean error less than 4 mmHg, standard deviation less than 7 mmHg (neonatal).
Heart rate meter range: 30-250 bpm.
Heart rate meter accuracy: ±3 bpm or ±3%, whichever is greater.
Scale: selectable °C or °F.
Range: 15°C to 45°C, 59°F to 113°F.
Accuracy: ±0.1°C to 45°C or ±0.2°F 59 to 113°F
Range: 4-200 pm.
Accuracy: ±2% or ± bpm, whichever is greater
Speed - 50, 25, 12., 6.25, 3.125 mm/sec ±5%
Type: 2 sealed lead acid batteries.
Voltage: 12 VDC nominal (@2.3A load).
Capacity: 2.3 AH each.
Run time: 1.5 hours from full charge of new battery at 25°C with SpO2, 1 NIBP every 15 minutes, no recording.
Recharge time: 16 hours.
Low battery warning: 11.6V, ±3.5%.
Low battery cut off: 11.0V ±2.8%
Qty - 1 Datascope Passport XG Multi-Parameter Monitor
Qty - 1 AMC 3-Lead ECG Cable CB-71320R (NEW)
Qty - 1 AMC 3-lead DIN to Snap Leadset LW-2090024/3A (NEW)
Qty - 1 Blood Pressure Cuff and Hose
Only items specified in the contents section are available with product.
Cosmetic - Very Good
Working - Very Good
Please contact us for any more information